Array of Ocean Bottom Tilt Current Meters: data from the EMSO-Azores observatory, 2018-2019
|Temporal extent||2018-08-17 -2019-06-25|
|Author(s)||Cannat Mathilde1, Wheeler Benjamin2, Chen Jie2, Sarradin Pierre-Marie3, Legrand Julien3|
|Affiliation(s)||1 : CNRS, IPGP, France
2 : IPGP, France
3 : Ifremer, France
|Keyword(s)||Mid-Atlantic Ridge, EMSO-Azores, EMSO, Hydrothermal Vents, Lucky Strike, Eiffel Tower, Crystal, Montsegur, current velocity, current direction, near seafloor seawater temperature, Time-series, MoMARSAT2018, MoMARSAT2019|
This dataset contains current data acquired between Auguest 2018 and June 2019 using 3 TCM3 Ocean Bottom Tilt Current Meters installed next to the Tour Eiffel, Montségur and Crystal hydrothermal vent sites.
The TCM-3 Ocean Bottom Tilt Current Meter from Lowell Instruments LLC (North Falmouth, MA, USA) measures current using the drag-tilt principle. The logger is buoyant and is anchored to the bottom via a short flexible tether. Drag from moving water tilts the logger in the direction of flow. The logger’s accelerometer and magnetometer channels are used to record the amount of tilt and direction of tilt (compass bearing). The array comprises 3 currentmeters, deployed near the Tour Eiffel, Montségur and Crystal hydrothermal vent sites. It is not connected to an energy node. The currentmeter's internal clocks are set to UTC time before deployment. Clock drift after recovery is not implemented in data but added as metadata in file Sensor Metadata.
|Utilisation||EMSO data are published without any warranty, express or implied. The user assumes all risk arising from his/her use of EMSO data. EMSO data are intended to be research-quality and include estimates of data quality and accuracy, but it is possible that these estimates or the data themselves contain errors. It is the sole responsibility of the user to assess if the data are appropriate for his/her use, and to interpret the data, data quality, and data accuracy accordingly. EMSO welcomes users to ask questions and report problems to the contact addresses listed in the data files or on the EMSO web page.|
Cruise: CANNAT Mathilde, LEGRAND Julien MOMARSAT2018 cruise, RV L'Atalante, http://dx.doi.org/10.17600/17000500
Cruise: SARRADIN Pierre-Marie, LEGRAND Julien (2019) MOMARSAT2019 cruise, RV Pourquoi pas ?, https://doi.org/10.17600/18001110
For deployments near selected hydrothermal vents at the EMSO-Azores observatory, the accelerometer and magnetometer are configured with a burst interval of 1 minute, a burst rate of 8 Hz and a burst duration of 15 seconds. Temperature are recorded with a period of 1 minute.
The specifications of the magnetometer, accelerometer and temperature probes are as follows:
Test and calibration A set of calibration coefficients for the magnetometer, accelerometer, and temperature channels are stored in the logger. The magnetometer and accelerometer coefficients are unique to each instrument while temperature coefficients are specific to each manufacturing lot. The calibration coefficients are applied upon recovery when the binary data is converted to text data. Data is provided as text, and not corrected for the time drift. Time drift is provided as metadata in Table_TCM3EMSO_Azores 2018-2019.xls.