Copy this text
Autonomous temperature probes data from the EMSO-Azores observatory, 2021-2022
This dataset contains temperature data acquired between june 2021 and june 2022 with an array of 32 autonomous temperature probes installed in smoker orifices, cracks or patches of diffuse hydrothermal venting. We used 7 types of probes (HTW, HTWN, HTNKE, LTNKE, LTW, LTI and LTGrad). Probes locations, acquisition parameters and clock drift values (UTC-timeprobe) are listed in the metadata file.
Data are provided for each deployed instrument as one or two text files, uncorrected for clock drift: temperature data.
Mid-Atlantic Ridge, EMSO-Azores, Hydrothermal Vents, Lucky Strike, Eiffel Tower, Capelinhos, Cimendef, Cypres, Montsegur, Sintra, South Crystal, White Castle, Y3, WHOI, NKE, iButton, LTGrad, Temperature, Time-series, MoMAR, MoMARSAT2021, MoMARSAT 2022
37.29211N, 37.28825S, -32.26407E, -32.28217W
- vent fluid temperature (°C)
- Regional node: Emso-Azores
- Site: Lucky Strike hydrothermal field
- coordinates: see EMSOAzores_auton_T_probes_2020_2021.docx
Cruise: MATABOS, Marjolaine and SARRAZIN Jozée, MOMARSAT2021 cruise, Atalante, https://doi.org/10.17600/18001296
Cruise: Cruise: SARRADIN Pierre-Marie and MATABOS, Marjolaine, MOMARSAT2022 cruise, Pourquoi Pas?, https://doi.org/ 10.17600/18001914
- Temperatures are recorded at frequencies between 1/30 s and 1/60 minutes.
High-temperature WHOI probes (HTW) contain of 2 J-type thermistors in their rod tip connected to 2 temperature data-loggers (named A & B), 2 data recordings are thus available for each probe. HTW probes record temperatures up to 415°C with an accuracy of ± 1.1°C, a resolution of 0.25°C at 375°C.
New high-temperature WHOI probes (HTWN) contain 1 temperature data-logger and 1 attitude logger to measure absolute orientation (tilt) and acceleration. The thermocouple in their rod tip can be J or T-Type. We mostly use J-Type thermocouples. These record temperatures up to 450°C, with an accuracy of ± 0.6°C, a resolution of 0,03°C.
High-temperature NKE probes (HTNKE) record temperatures up to 450°C, with an accuracy of ±0.5°C, and a resolution of 0.1°C.
Low-temperature WHOI probes (LTW) record temperatures up to 125°C, with an accuracy of ±0.22°C, and a resolution of 0.025°C.
Low-temperature NKE probes (LTNKE) record temperatures up to 100°C, with an accuracy of ±0.5°C, and a resolution of 0.12°C at 60°C.
IPGP LT probe (LTI). This probe is a prototype with a titanium housing a xxx thermocouple and a OM-CP-RTDTEMP101A (Omega) logger. It record temperatures up to x°C, with an accuracy of ±x°C, and a resolution of x°C.
LTGrad probes comprise 2 LTiButton probes separated by 7 cm and attached to a 25 to 30 cm-long PVC support. Low-temperature iButton probes (LTiButton) record temperatures up to 85°C, with an accuracy of ±0.5°C, and a resolution of ±0.5°C.
Test and calibration
All probes are synchronized to UTC time before deployment. Temperature readings of WHOI probes are checked with an oven at IPGP whenever the probe is not redeployed... which is not frequent because we do not have many spares. NKE probes are sent back to maker every 3-4 years for battery change and calibration. Clock drift at recovery is listed in the metadata file. EMSOAzores_auton_T_probes_2021_2022.docx
- Real time data transmission : No
- Data storage in the Communication and storage unit (COSTOF2): No
- Data storage in the instrument: Full
|File||Size||Format||Processing||Access||end of embargo|
|10 Mo||TEXT||Raw data||2024-10-01|